## You are here: Home > News > 201-12 What is the difference of particle size measurement between fisher sub sieve sizer (FSSS) and laser particle size analyzer (LPSA)?

## 201-12 What is the difference of particle size measurement between fisher sub sieve sizer (FSSS) and laser particle size analyzer (LPSA)?

201-12 what is the difference of particle size measurement between fisher sub sieve sizer (FSSS) and laser particle size analyzer (LPSA)?

The particle size result obtained by fisher sub sieve sizer is the average particle size value DSV, which is somewhat similar to the particle size result D(3,2) obtained by the laser particle size analyzer. However the average particle size meter actually carried out this test, while the laser particle size meter calculated D(3,2) on the premise that the particles were all circular after particle size distribution measurement. that is to say, the difference between circular particles is small, and the difference between irregular particles is large. In terms of measurement result, there is no comparability between the fisher sub sieve sizer and laser particle size analyzer. Because the principle of measurement is different. But in terms of the measurement of equicircular particles, both of the instrument can get satisfying test result.

Laser particle size analyzer, because the sample cannot one hundred percent dispersed into a particle, the actual measurement of the sample is a small group of particles, the particle size will generally be larger. The adsorption method is converted by the adsorption amount, because the particles must be agglomerated, so the results will be larger than the actual particle size. According to experience, the result of laser particle size meter is much larger. For example, AS-2011 Laser particle Size Analyzer and HMK-22 Fisher Sub Sieve Sizer, The actual sample particle size is 0.3-0.4 micron on average (electron microscope), fisher sub sieve sizer usually can measure from 1.0 to 1.5 microns, laser particle size analyzer can usually measure from 2 to 3 microns. Generally speaking, the laser particle size analyzer id greatly affected by external conditions, such as the choice and dose of dispersant, ultrasonic dispersion intensity and time, etc., the repeatability is not as good as that of fisher sub sieve sizer. The laser particle size is not used as a parameter in the process in industry.

Tags: air permeability,air permeability diameter,air permeability particle sizing,air permeability principle,air permeability theory,ASTM B 330,ASTM Method B 330,ASTM Standard B330-12,average diameter,average material particle size,average particle diameter,average particle size,Average Particle Size Analyzer,average particle size d50,average particle size d50 definition,average particle size diameter calculation,average particle size of,average powder size,average size,average specific surface area,buy fisher sub sieve sizer,Fisher FSSS,Fisher Model 95 Fisher Sub-Sieve Sizer,fisher model 95 parts,fisher model 95 sub-seive sizer,Fisher Model 95 Sub-Sieve Sizer,Fisher Model 95 Sub-Sieve Sizer (FSSS),Fisher number,fisher number (FSSS) particle sizing system,Fisher number of Metal Powders,fisher number particle sizing system,fisher number results report,fisher particle size,fisher replacement parts for sub-sieve sizer,fisher scientific sub sieve sizer,fisher scientific sub sieve sizer model 95,fisher sizer,fisher sub sieve analyzer,fisher sub sieve size,fisher sub sieve sizer accessories,fisher sub sieve sizer extractor rod,fisher sub sieve sizer filter paper disc,fisher sub sieve sizer fsss,fisher sub sieve sizer instruction manual,fisher sub sieve sizer manual,fisher sub sieve sizer model 95,fisher sub sieve sizer plastic funnel,fisher sub sieve sizer porous plugs,fisher sub sieve sizer products,fisher sub sieve sizer rubber stopper,fisher sub sieve sizer sale,fisher sub sieve sizer sample pressure calibrator,fisher sub sieve sizer sample tube,fisher sub sieve sizer sample tube holder,fisher sub sieve sizer sub sieve sizer calibrator,fisher sub-sieve,Fisher Sub-Sieve Sizer,fisher subsieve,fisher subsieve analysis,Fisher Subsieve Analysis of Metal Powders,Fisher Subsieve Sizer,FS number,FSSS,fsss instrument,fsss particle size,HEL SAS,HEL Subsieve AutoSizer,iso 10070,mean particle size,measure average particle size,measure particle sizes,measures average particle size,Measuring Particle Size Average,metal particles,metal powder sizing technique,metal powders,Micromeritics Sub-Sieve AutoSizer,MODEL 95,model 95 fisher subsieve sizer,particle size,particle size analyzer price,particle size and surface area,particle size average,Particle Size Average of metal powders,particle size surface area,particle size vs surface area,powder size,powder sizing by air permeability,range of fisher sub sieve sizer,refractory metals and compounds fisher number,SAS,specific surface area and particle size,specific surface area particle size,Standard Test Methods for Estimating Average Particle Size of Metal Powders and Related Compounds Using Air Permeability,sub sieve sizer,volume average particle diameter,volume average particle size,volume average particle size and diameter analyzer